A New Tool for the Automated Sample Preparation of Whole Blood Samples by LC-MS using a Commercial Autosampler
Poster Jul 06, 2017
Christian Berchtold, Guenter Boehm, Renée Falsia, Thomas Preiswerk, Götz Schlotterbeck
- The new LCMS-P 80 Tool is capable of pipetting whole blood with normal hematocrit levels (30-50%) directly from primary tubes with good accuracy and precision, applying one average correction factor. For high hematocrit levels a different correction factor should be applied.
- Wash procedures were established to minimize carryover of analytes. A procedure to minimize the deposition of blood on system surfaces proofed successful.
- The system consisting of a commercial PAL RTC autosampler, connected to an Agilent LC-MS system can perform the complete sample prep and LC-MS analysis for whole blood samples from primary tube. The performance of the system was tested with several typical compounds in therapeutically relevant concentrations.
- Future work is aimed at evaluating the capability of the sensor to reliably detect certain error conditions.
Analysis of Extract Drying Criteria for Oil & Grease Method 1664A/BPoster
Analysis of Extract Drying Criteria for Oil & Grease Method 1664A/BREAD MORE
Designing a Model to Explore Tau's Unfolded Protein ResponsePoster
The purpose of this research is to design a cell model in which ER stress caused by tau accumulation can be generated, and then investigated for changes in different ER stress-associated proteins.READ MORE
Fundamentals and Comparisons for Organic Sample Extract EvaporationPoster
Sample preparation is a key step in the analysis process
Parameters for evaporation and their impact on analysis have been discussed
Improvements in matching the sample to the evaporation device characteristics can help reduce variability and improve recovery
Examples for choosing a system based on sample volume, types of analytes, sample load, and initial investment considerations