PAL SPME Arrow for the determination of volatile organic compounds in water by GC-MS after headspace-solid-phase microextraction (HS-SPME) according to German standard method DIN 38 407-41
Poster May 24, 2017
Andreas Kremser (1), Guenter Boehm (2)
The German standard DIN 38 407-41 describes a method for the analysis of volatile compounds from aqueous samples by headspace-solid-phase microextraction (HS-SPME). Analytes are volatile, organic compounds (VOCs), fuel constituents (BTEX) and methyl-tert.-butylether (MTBE). The method allows for the detection of such contaminants in drinking-, ground- and surface waters by gas chromatography coupled to mass spectrometry (GC-MS). With the small sample vol¬umes of 10 mL, the working range of the original method is 0.01 to 100 μg/L, with the lowest reported detection limits being in the range of approx. 10 ng/L.
PAL SPME Arrow is a suitable tool for fulfillment of the German standard method DIN 38 407-41 for head¬space-solid-phase microextraction of volatile analytes from aqueous samples. Obtained detection limits with the novel device are at least one order of magnitude better than the values that were reported for the classical SPME fiber. Meth¬od repeatability and linearity are on par for both techniques. In addition, the improved mechanical reliability of PAL SPME Arrow can be expected to benefit the overall method stability over prolonged, automated measurement series.
Analysis of Extract Drying Criteria for Oil & Grease Method 1664A/BPoster
Analysis of Extract Drying Criteria for Oil & Grease Method 1664A/BREAD MORE
New Biotransformation Prediction Engine Integrated into a Metabolite Identification SolutionPoster
Here we present a new prediction algorithm that determines the likelihood of biotransformation reactions, and subsequent metabolite identification, within an automated processing routine.READ MORE
Fundamentals and Comparisons for Organic Sample Extract EvaporationPoster
Sample preparation is a key step in the analysis process
Parameters for evaporation and their impact on analysis have been discussed
Improvements in matching the sample to the evaporation device characteristics can help reduce variability and improve recovery
Examples for choosing a system based on sample volume, types of analytes, sample load, and initial investment considerations