Environmental Studies Using the Tecan QC Kit and an Automated Liquid Handling System
Poster Feb 10, 2015
George Rodrigues, Emily Avis, Alf Price, Verena Gerber
Liquid handler performance can be optimized by adjusting system operating parameters, but the effect of environmental conditions on liquid handling quality is often underappreciated.
The study described in this poster illustrates the effect of two environmental factors (air temperature and humidity) on the volumetric performance of a Tecan Freedom EVO ® pipetting robot, equipped with a liquid LiHa (Liquid Handling Arm). Tecan recommends that the EVO be installed in a controlled environment. However, some users choose to operate in less controlled conditions.
This study uses a two factor design (temperature and humidity) to investigate volumetric performance (precision and trueness) at environmental conditions from 15 °C to 30 °C and 30% to 80% relative humidity. Nine environmental conditions were tested, the midpoint, four corner cases and four edge cases. Replicate testing was performed at the midpoint and corner cases.
The Tecan QC Kit, which is based on Artel’s ratiometric photometry, was used to make measurements at each set of environmental conditions. The precision (%CV), and trueness (%SE) is reported for each environmental condition using default liquid classes.
A New Tool for the Automated Sample Preparation of Whole Blood Samples by LC-MS using a Commercial AutosamplerPoster
Automated sample preparation reduces the costs per sample and minimizes sample handling errors. Usually expensive and highly specialized pipetting robots are used. However, most of these systems are not designed with a direct interface for LC-MS applications. In addition common pipetting systems are not optimized for smaller scale sample series. Here we present a new tool for liquid handling of whole blood samples and direct sample injection.READ MORE
Characterization of Heavily Contaminated Sites using NMR SpectroscopyPoster
NMR spectroscopy provides a unique analytical approach to present a more complete an unbiased understanding of the composition of NAPLs at many contaminated sites.READ MORE
PAL SPME Arrow for the determination of volatile organic compounds in water by GC-MS after headspace-solid-phase microextraction (HS-SPME) according to German standard method DIN 38 407-41Poster
Robust and sensitive determination of volatile organic compounds in water by headspace solid phase microextraction (HS-SPME) according to German standard method DIN 38 407-41READ MORE