Minimizing Carry-over for High Throughput Analysis
Poster Aug 03, 2015
Christian Berchtold1, Reto Bolliger2, Guenter Boehm2, Götz Schlotterbeck1
A LC-MS system consisting of an Agilent 1260 Infinity HPLC and Agilent 6120 Quadrupol LC-MS equipped with a multimode ion source (set on ESI mode) was used. A PAL RTC sample preparation system with a LC/MS-Tool, high pressure injection valve (VICI C72VC-6676D-CTC) with a 2 µL loop was used. For separation a high throughput gradient HPLC method of 2 minutes cycle time was established using an Agilent Zorbax SB C18 Column (3 µm, 2.1x50mm) for chlorhexidine injections.
The PAL RTC offers five major wash tasks for carry-over minimization, such as post clean solvent 1 and 2, valve clean solvent 1 and 2, and stator wash. Solvent 2 is chosen according to analyte and matrix solvability, since it is used as the major wash step directly after each run. In contrast, solvent 1 is used to maintain reproducible conditions for subsequent analyses (therefore usually the eluent at starting conditions is used). The parameters of each wash step have been systematically screened using an on-column injection of chlorhexidine followed by fresh blanks. For chlorhexidine a mixture of water/methanol/acetonitrile/2-propanol containing 1% formic acid (for solvent 2) resulted in minimal carry-over.. A total of 5 strokes and 200 µL solvent 2 and two strokes and 90 µL solvent 1 for post clean and valve clean have been used respectively. Usinging this conditions a carryover of less than 20 ppm has been observed usuing wash cycles of less than 2 minutes for chlorhexidin. An optimized auto sampler wash protocol was developed and general guidelines for method development for HT LC-MS applications were established. The coloumn has been identified as the main source of carry-over.
Sport Doping Screening in Biological Matrices by Multi-Dimensional LC-QToFPoster
This work evaluated the performance of 2D LC variant using a QToF setup instead of a triple quadrupole mass spectrometer for the analysis of drug of abuse in urine targeting low and sub ppb level.READ MORE
Analysis of Doping and Forensic Drugs in Urine Using High-Resolution GC/Q-TOFPoster
In this study, we are examining the potential for high resolution accurate mass 7250 GC/Q-TOF equipped with low energy EI source, for both quantitative and screening aspects of doping control and forensic drugs applications.READ MORE
Automation of Sample Preparation for Metabolomic Analysis Using Robotic PlatformPoster
We developed an automated sample preparation protocol based on a robotic platform PAL RTC (CTC Analytics AG, Zwingen Switzerland), which represent a modified Bligh and Dyer method producing samples for both hydrophilic metabolomics using GC-MS and lipidomics using SFC-MS simultaneously.READ MORE